to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||18 December 2012|
|PDF File Size:||6.13 Mb|
|ePub File Size:||10.25 Mb|
|Price:||Free* [*Free Regsitration Required]|
Repeat the measurement using a new test strip. Performing a measurement Start measurement by closing the Display of measurement results measurement chamber flap This column contains the illustration of a display This column contains information relating to the screen. Table of Contents Add to my manuals Add.
Open the battery compartment by slightly pressing the tab towards the center of the instrument. Flap open displayed after instrument has been powered on The measurement chamber flap is not closed properly.
Instrument settings If the displayed setting is correct e. Quality Control Recommendation Quality control recommendation Quality control recommendation To ensure that the instrument is functioning properly, you should regularly use two levels of control solutions for a function check.
Preparing To Perform A Control Test Quality control recommendation Preparing to perform a control test Get the test strip container for the required measurement e. Get the proper control solutions for the test strips.
Page 63 Performing a measurement Hold the test strip with thumb and index finger so that the sample application area is facing upwards. Performing a measurement The flashing drop symbol above the strip symbol now instructs you to apply the blood.
Performing a measurement Check that the whole display is functioning correctly to prevent misinterpretations due to defective display elements. Always use test strips before their expiration date has passed.
Cobas Accutrend Plus User Manual
The Accutrend Plus instrument Power To save power, the instrument powers itself off after 2 minutes unless a button is pressed or a new test strip is inserted. Solution Remove and discard the strip. This starts the actual measurement. Set date, time and beeper. If the entire memory area of a test parameter is empty, two dashes — — will appear in the display.
Performing a measurement Code strips The code strip provides the instrument with important information on the specific properties of the respective test strip.
Insert code strip can also be done directly before performing the measurement.
Press the Set button to save the setting. Wash your hands with warm water. In addition, you can flag measurements carried out with control liquids as controls. Display Of Results Quality control recommendation Display of results When measurement is finished, the result is displayed.
Quality control recommendation Applying the control solution 10 Apply a large free hanging drop of either the glucose or cholesterol low level control solution directly from the bottle to the test strip. Blood can be applied either in the instrument or outside the instrument. The next time the Set button is pressed, the minutes can be set again with the M button.
Instrument Settings Operating the instrument Instrument settings Brief overview of the instrument settings The following table provides an overview of the available settings. Insert the test strip into the test strip guide up to the stop. Refer to safety-related notes in the instruc- tions for use accompanying this product.
Consult instructions for use. The mannual area must be completely covered with blood example A kanual, otherwise incorrect values may be obtained. Further Information Accutrend Cholesterol Control Control solution to be used with Accutrend Cholesterol test pljs Accutrend Plus instrument kit Meter kit Product majual Please read the insert packaged with the test strips or control solutions for detailed information on product data Power The Accutrend Plus instrument Power To save power, the instrument powers itself off after 2 minutes unless a button is pressed or a new test strip is inserted.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Integrated Control Functions Introduction Integrated control functions The Accutrend Plus instrument has numerous integrated or available control functions, including the following: Page 93 Memory After the usual display test has been performed when switching on the instrument, the last stored measured value is displayed.
Page 79 Performing a measurement Assigning additional information to a measurement: Operating the instrument Instrument settings Brief overview of the instrument settings The following table provides an overview of the available settings. Setting The Date Format Instrument settings Setting the date format In the first accutremd, you set the date format the entire date flashes. Brief Overview Of The Steps To Be Carried Out Performing a measurement Brief overview of the steps to be carried out Code the instrument once Insert the test strip Apply blood per test strip lot Page 46 Performing a measurement Mnaual measurement by closing the Display of measurement results measurement chamber flap Caution, consult accompanying documents.
Medical staff and other persons using the Accutrend Plus instrument kanual perform measurements for more than one patient must be aware that any object coming into contact with human blood is a potential source of infection. Error Messages Error messages Error messages In certain circumstances error messages may appear on your display.
When the test strip reaches the correct position, you will hear two beeps short — The flap firmly locks into place when it reaches a vertical position. Page 87 Quality control recommendation Open the measurement chamber flap. Performing a measurement Switching the code display You can switch between the display of the stored code numbers for information purposes.
Unknown barcode The barcode of test strips or code strips could not be read. The display then automatically switches to the mode for setting the current day. Make sure that your assignments are unique and reproducible. Hold the test strip with thumb and index finger so that the printed measurement parameter is facing upwards. Page 35 Instrument settings If the displayed setting is correct e.
Setting The Time Instrument settings Setting the time First enter the current mahual and then the minutes. Cleaning The Optical Measuring System Cleaning Cleaning the manul measuring system Clean the easily accessible areas of the optical measuring system under the strip guide with a lint-free pad, a moistened cotton ball, or a moistened cotton swab.