The Nonin® Onyx Model Finger Pulse Oximeter is a small, lightweight, The use of accessories other than those specified in these instructions may result . View and Download Nonin Onyx instructions for use online. Finger Pulse Oximeter. Onyx Medical Equipment pdf manual download. Medical Equipment Nonin PalmSAT Operator’s Manual. (35 pages). Medical Equipment Nonin Onyx Instructions For Use. Finger pulse oximeter (5.
|Published (Last):||10 June 2012|
|PDF File Size:||15.27 Mb|
|ePub File Size:||11.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
Contents Continued Accuracy Testing Installing Batteries in the Verify that all visible indicators illuminate during the startup initialization sequence. If any indicator is not lit, do not use the device.
Download service manual for nonin onyx | Diigo Groups
Contact Nonin Technical Service for assistance. The presence of a defibrillator may interfere with the performance of this device. Do not attempt to open the case or repair the electronics. Opening the case may damage the device and void the warranty.
Any sign or evidence of opening the system, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the system, shall void the warranty in its entirety. Guide to Symbols Guide to Symbols This table describes the symbols that are found on the Models and M and in this manual.
Mamual Instructions for Use. Follow Instructions for Use. This 3-digit light-emitting diode LED display shows the current oxygen saturation percentage. Contact the carrier immediately if the shipping carton is damaged. Before changing the batteries, make sure the device is off and the sensor is not applied to a digit. Using the and M Pulse Oximeters Figure 2: Installing Batteries in the When batteries are critically low, the digital displays will go blank, and the Pulse Quality display will blink amber or red, but not green.
After 10 minutes at critically low battery capacity, the pulse oximeter will shut off automatically. Connecting the Sensor Connect the pulse oximeter sensor with the Nonin logo facing up to the top of the device as shown.
Ensure that the sensor is firmly plugged in. Set the calendar and clock.
Memory download function M only. Care and Maintenance Care and Maintenance Clean the device separately from the sensors.
For instructions on cleaning pulse oximeter sensors, refer to the respective sensor instructions for use. Do not autoclave or immerse the device or sensors in liquid. Do not expose the device or components to excessive moisture or liquids. Visual Indicators Visual Indicators The following table describes conditions and visual indicators. Replacing the batteries before 3 minutes have elapsed most likely will result in corrupt data. Communications Communications Serial Output Models and M provide real-time data output capability via the pulse oximeter sensor connector a 9-pin Sub-D connector.
The pulse oximeter sensor connector pin assignments are listed below. Nonin does not recommend field repair of Models and M. The circuit board in Models and M is a multi-layer board using very narrow traces. Due to the very small trace size, extreme care must be used when replacing components to prevent permanent, non-repairable damage to the circuit board.
Model or M, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the Model or M, shall void the warranty in its entirety. All non-warranty work shall be done according to Nonin standard rates and charges in effect at the time of delivery to Nonin. For more information about Nonin parts and accessories: The batteries are installed Verify battery orientation, incorrectly.
A metal contact in the battery Contact Nonin Technical compartment is missing or Service. A non-compatible sensor is being Replace the sensor with a used. The ECG monitor may not be Assess the patient.
A functional tester cannot be used to assess the accuracy of a pulse oximeter monitor or sensor. All parts and accessories connected to the serial port of this device must be certified according to at least IEC Standard EN or UL for data- processing equipment. Technical Information Table 4: The user of this device should ensure that it is used in such an environment.
Technical Information Table 5: Portable and mobile RF communications equipment should be used no closer to any part of the device, including cables, than the recommended separation distance calculated from the equation applicable to the frequency of the transmitter.
Technical Information Table 6: Recommended Separation Distances This table details the recommended separation distances between portable and mobile RF communications equipment and this device. This device is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. Specific to this example: Testing Summary accuracy, and low perfusion testing was conducted by Nonin Medical, Inc.
Accuracy Testing During motion and no-motion conditions at an independent research laboratory, SpO Page of 33 Go. Electromagnetic Immunity Page 30 – Table 5.
Guidance and Manufacturer’s Dec Table of Contents Add to my manuals Add. Wrist-worn pulse oximeter with bluetooth wireless technology 51 pages.
Page 2 Authorized EC Representative: Page 4 Contents Continued Accuracy Testing Page 5 Figures Figure 1. Page 6 Tables Table 1. Cautions Verify that all visible indicators illuminate during the startup initialization sequence. Page 9 Do not attempt to open the case or repair the electronics.
Nonin Onyx 9500 Instruction Manual
Using The And m Pulse Oximeters Contact the carrier immediately if the shipping carton is damaged. With Aa Batteries 6 weeks. Memory Functions m Only 3 minutes after removing the batteries. Communications Communications Communications Serial Output Models and M provide real-time data output capability via the pulse oximeter sensor connector a 9-pin Sub-D connector.
Page 23 Model or M, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the Model or M, shall void the warranty in its entirety. Parts And Accessories For more information about Nonin parts and accessories: Troubleshooting The batteries are installed Verify battery orientation, incorrectly. Page 26 A non-compatible sensor is being Replace the sensor with a used. Page 27 Sensor failure. Electromagnetic Immunity Technical Information Table 4: Testing Summary Specific to this example: This manual also for: Comments to this Manuals Your Name.
Enter text from picture: By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it.
Don’t show me this message again. Got it, continue to print. Don’t have an account?